The Scanning PIB Locators step
should be completed.
Consider using the Advanced Settings before scanning an
Spray any translucent or non-optical scan material with a CAD spray in order for the
scanner to accurately detect these materials. Failure to do so can result in a lack
of data or less accurate data.
From the Camera View, set the
blue square to cover all relevant model information needed to properly design
Consider rotating the mouse wheel to increase the size of the blue
rectangle to fully cover the model. You can add points to the rectangle
by clicking on the central line running through the rectangle.
Position the blue circles over the reference circles of the holder.
If a warning is displayed indicating that Height validation failed,
adjust the height of the table until the red column changes to
Note: 1G : it is sometimes acceptable to ignore the height validation warning and proceed
with scanning. A scan of the area will be added to the initial
Make sure the blue circles and square do not need to be
When scanning is completed, click Ok.
Turn the Procera® Implant Bridge Framework upside down.
Repeat steps 1 to 4.
During scanning, complete the information on dentist, technician, identifier
From the restoration list, select a material and shade for each
A situation is created with the scan. The situation is sent to the NobelProcera Crown and
where the bridge can be designed.